Spectral Line Observations
The ceduna correlator is corr-ced and can be reset from the internet power switch.
The DAS is controlled using the xdos program on ceduna (the oper login).
The local oscillators can be set using rs.tcl on ceduna (the oper login).
To start the correlator
To kill the correlator (should be done at the end of observations and not left running indefinitely)
The levels into the DAS are controlled by some dial attenuators, so if the levels aren’t right you need someone there in person to change them.
If you are having problems getting the correlator to program correctly then you can check the status as it happens by